by SAE International in United States
Sound intensity measurement techniques that used a two-microphone configuration, were first developed in the late 1970s. Originally, the focus was on improving precision during testing or post-processing. However, with the advent of modern, sophisticated equipment, the focus has shifted to the apparatus. Availability of phase-matched microphones has made post-test correction obsolete as the microphones eliminate a majority of the errors before the data is even collected. This accuracy, however, comes at a cost, as phase-matched microphones are highly priced. This paper discusses employing the method of improving post-processing precision, using inexpensive, current equipment. The phase error of the system is corrected using a simple calibration technique and a handheld phase calibrator that is similar to the one used for amplitude calibration of microphones. The intensity probe and calibrator is manufactured using rapid prototyping and the executable software that goes with the probe is designed in NI LabVIEW. The entire setup uses inexpensive parts to lower the cost and modern software to compensate for the errors due to these parts. The design of the probe and…
This content contains downloadable datasets
Annotation ability available
A list of downloadable items.
Please Wait. Loading Available Formats...
This modal will allow the user to download citation information or export specific fields of data into one of three file formats: Excel, comma-separated values, or text.
Sorry, this document is already saved to a project folder.
Share this content via social media.
Share Via Email
Email sent successfully!
Use this modal to save a search.
The current search and criteria have already been saved. Please modify your criteria to save as a new search.
Request Sent successfully!
Search Tips: Search Operators:AND, OR, NOT; Wild Cards; Using Quotes
SAE MOBILUS® Search Tips
Please note: All terms not contained within quotes, will be evaluated using the OR operator. This means that your search results will contain all results that contain ANY of the entered terms.
AND, OR, NOT
SAE MOBILUS supports the use of the AND, OR and NOT BOOLEAN operators ONLY function with the Advanced Search.
Using the asterisk, ( * ) allows you to search for a partial word. For example, entering a keyword search of aero* will search for any word beginning with aero.
Using the question mark, ( ? ) allows you to search for a word with a single character wildcard. For example, searching for t?re would match both tire and tyre
Words in double quotes " " are treated as a phrase. Conducting a search for "heat model" will return all results in which the exact phrase "heat model" is found, while "heat modeling" would NOT be included.
A term combined with a phrase will be joined with an OR operator.
Searching for Standards
When searching for a standard please use the entire Document Number assigned, which includes the abbreviation. A space can be used, or omitted between the abbreviation and number.