This content is not included in your SAE MOBILUS subscription, or you are not logged in.
Video Based Light Measurement Techniques
- Ground Vehicle Standard
Published July 24, 2014 by SAE International in United States
Traditional methods of photometry rely on the use of a goniometer to rotate the test item around two axes at right angles. This method is satisfactory for most situations but has certain disadvantages:
Point-by-point measurements with a goniometer may be slow. With more advanced requirements, particularly for headlamps, where the entire beam pattern is of concern, isocandela measurements are becoming increasingly needed. Such testing can be very time consuming.
For production quality assurance, the speed of a goniometer may not allow testing to keep pace with the production line if a large quantity of lamps must be sampled.
High Intensity Discharge (HID) lamps are becoming commonly used. Such lamps are orientation sensitive, changing in both lumen output and intensity distribution when tilted. This can introduce significant inaccuracies in test results when testing is performed using a goniometer.
There is a need for alternative test techniques which can achieve very high speed data acquisition, the capture of full isocandela distribution, and the elimination of lamp tilting. This SAE Informational Report describes fundamentals of video-based testing to address these concerns. Further information is required to provide all details needed to set up a laboratory using these techniques.
With the video-based system, the lamp is fixed in position and aimed at a receiving screen. A camera, of particular type and grade, views the screen and is able to perform measurement of the reflected light (luminance) at each point on the screen. Proper calibration of the system can provide the luminous intensity distribution of the light under test from the measured luminance values.
Numerous factors must be taken into account to produce an accurate measurement system. Under controlled conditions, it has been shown that the system can produce measurement of absolute intensity distributions with good accuracy. (Reference 2.1.1 and 2.1.2.)
The great advantage of this form of photometry is its extremely high speed, allowing the collection of isocandela diagrams in less than one minute. There are some accompanying disadvantages which are discussed later.
This document describes the basic techniques and the controls which are necessary for accuracy.
Comments are also provided comparing video based technology to conventional goniometer methods.
This document will be Stabilized and kept only for historical reference only.
|Technical Paper||Production Code Generation for Safety-Critical Systems|
|Technical Paper||The Development of a Semi-Automated Radio Test System|
|Technical Paper||Embedded Software Engineering in automotive and Truck Electronics|
Data Sets - Support Documents
No Datasets Available
|960919||Advances in Measurement Technology for Vehicle Lighting Systems|
|970231||Accuracy Analysis of Video-Based Photometry|
* Redlines comparisons are available for those standards listed in the Revision History that contain a radio button. A redline comparison of the current version against a revision is accomplished by selecting the radio button next to the standard and then selecting 'compare'. At this time, Redline versions only exist for some AMS standards. SAE will continue to add redline versioning with ongoing updates to SAE MOBILUS.