This content is not included in your SAE MOBILUS subscription, or you are not logged in.
Hybrid III Family Chest Potentiometer Calibration Procedure
- Ground Vehicle Standard
Published October 07, 2022 by SAE International in United States
This procedure establishes a recommended practice for establishing the sensitivity of the chest displacement potentiometer assembly used in the Hybrid III family of Anthropomorphic Test Devices (ATDs, or crash dummies). This potentiometer assembly is used in the Hybrid III family to measure the linear displacement of the sternum relative to the spine (referred to as chest compression). An inherent nonlinearity exists in this measurement because a rotary potentiometer is being used to measure a generally linear displacement. As the chest cavity is compressed the potentiometer rotates, however the relationship between the compression and the potentiometer rotation (and voltage output) is nonlinear.
Crash testing facilities have in the past used a variety of techniques to calibrate the chest potentiometer, that is to establish a sensitivity value (mm/(volt/volt) or mm/(mvolt/volt)). These sensitivity values are used to convert recorded voltage measurements to engineering units, in this case chest compression in mm. Some of these techniques intended to correct for the nonlinearity and others did not. Of those that did correct for the nonlinearity, there was a variation in techniques used. This variation in calibration procedures was in part identified by the SAE Dummy Testing Equipment Committee (DTEC) and led to overall variability in chest compression measurements between laboratories.
The intent of this recommended practice is to minimize the variations in chest deflection measurements between crash testing laboratories. Before this procedure was written, a round robin showed variations for the Small Female of 10% among eight labs for the chest pot sensitivity value. A follow-up round robin of this procedure showed a worst case variation of 2.7% among 10 labs, with a standard deviation of 0.9%. The initial version of SAE J2517 released in May 2000 attempted to fix this problem by recommending a two-point calibration which was not intended to correct for the nonlinearity (which, for example, is as large as 3% for the Small Female but is small near the peak). It also did not require the measurement of a starting position of the potentiometer before each crash test, thus it did not correct for the difference in starting chest geometry between a subject dummy and its design intent. It was intended to be a simple and reproducible calibration procedure which crash test facilities could easily adopt with little or no modifications to their facilities. In practice, most laboratories did not adopt the procedure since it did not correct for the nonlinearities.
Recent attempts to reduce dummy-to-dummy and lab-to-lab variations at lower deflection levels (around 25 mm) have renewed interest in adopting a calibration procedure to correct for the nonlinearity of the measurement system. This current revision of this procedure uses a multipoint calibration with a third order regression to correct for the nonlinearities of the system with a standardized method. It requires changes in the calibration method of the transducer, the data collection procedures when used in a dummy, and the processing procedures after test data is collected. Following this standardized methodology will minimize linearity errors as well as lab-to-lab variations.
SAE J2517 has been revised to replace references to Engineering Aids 25 and 26 with references to SAE J2860 and SAE J2862 where applicable. Also added references to SAE J2858.
Data Sets - Support Documents
No Datasets Available
Dummy Testing and Equipment Committee
The Dummy Testing Equipment Committee (DTEC) is a group of experts with experience in the area of certification and handling of Anthropomorphic Test Devices (ATD), more commonly known as crash test dummies, and equipment related to ATD. The DTEC seeks to standardize the use and certification of ATD so that test data are both repeatable and reproducible.
|EngineeringAid2||This document is not part of the subscrption.|
|FMVSSNo.572||This document is not part of the subscrption.|
|J211/1_201403||Instrumentation for Impact Test - Part 1 - Electronic Instrumentation|
|J2570_201911||Performance Specifications for Anthropomorphic Test Device Transducers|
|J2706||This document is not part of the subscrption.|
|J2706_201306||Hybrid III Six-Year-Old Child Dummy User's Manual|
|J2857_202301||This document is not part of the subscrption.|
* Redlines comparisons are available for those standards listed in the Revision History that contain a radio button. A redline comparison of the current version against a revision is accomplished by selecting the radio button next to the standard and then selecting 'compare'. At this time, Redline versions only exist for some AMS standards. SAE will continue to add redline versioning with ongoing updates to SAE MOBILUS.