This content is not included in your SAE MOBILUS subscription, or you are not logged in.
Influence to the PN Emissions in Calibration Procedure for Portable and Stationary Solid Particle Number Measurement Systems
ISSN: 0148-7191, e-ISSN: 2688-3627
Published December 19, 2019 by SAE International in United States
This content contains downloadable datasetsAnnotation ability available
The Real Diving Emissions (RDE) regulation has been introduced since September, 2017 by utilizing the Portable Emissions Measurement System (PEMS). For the PEMS for the solid Particle Number (PN) measurement (PN-PEMS), the validation tests are required by comparing to the stationary PN measurement system on a chassis dynamometer prior to the on-road emissions testing. However, there are some cases that the emission results of PN-PEMS have big difference for that of the PMP system as the PN-PEMS does not have the same system configuration and calibration procedures as a PMP system. In this paper, the influence of the calibration procedure to the PN emissions results was observed by applying the calibration procedure of the PN-PEMS to the PMP system. The current systems configurations for PMP system and PN-PEMS, and the differences of them were described. And, the calibration procedure of the PN-PEMS was applied to the PMP system to adjust the system detection efficiency at 23 nm. Finally, the influence of the calibration protocol was investigated by compared the PN emissions of PMP system and the on PN-PEMS.
CitationKondo, K., Kojima, K., Kusaka, T., and Otsuki, Y., "Influence to the PN Emissions in Calibration Procedure for Portable and Stationary Solid Particle Number Measurement Systems," SAE Technical Paper 2019-01-2196, 2019, https://doi.org/10.4271/2019-01-2196.
Data Sets - Support Documents
|[Unnamed Dataset 1]|
|[Unnamed Dataset 2]|
|[Unnamed Dataset 3]|
- Weiss, M., Bonnel, P., Hummel, R., Manfredi, U. et al. “Analyzing on-road emissions of light-duty vehicles with portable emission measurement systems,” JRC Scientific and Technical Reports EUR 24697 EN, 2011.
- Hu, S., Zhang, S., Sardar, S., Chen, S. et al., “Evaluation of Gravimetric Method to Measure Light-Duty Vehicle Particulate Matter Emissions at Levels below One Milligram per Mile (1 mg/mile),” SAE Technical Paper 2014-01-1571, 2014, doi:10.4271/2014-01-1571.
- Chase, R.E., Duszkiewicz, G.J., Richert, J.F.O., Lewis, D. et al., “PM Measurement Artifact: Organic Vapor Deposition on Different Filter Media,” SAE Technical Paper 2004-01-0967, 2004, doi:10.4271/2004-01-0967.
- Zervas, E., Dorlhene, P., Forti, L., Perrin, C. et al., “Inter-laboratory Test of Exhaust PM using ELPI,” Aerosol Sci. Technol. 39:333–346, 2005.
- Giechaskiel, B., Dilara, B., and Andersson, J., “Particle Measurement Programme (PMP) Light-Duty Inter-Laboratory Exercise: Repeatability and Reproducibility of the Particle Number Method,” Aerosol Sci. Technol. 42:7, 528-543, 2008, doi:10.1080/02786820802220241.
- Otsuki, Y., Nakamura, H., Arai, M., and Xu, M., “The methodologies and instruments of vehicle particulate emission measurement for current and future legislative regulations,” Meas. Sci. Technol., 26(9), 092002, 2015, doi:10.1088/0957-0233/26/9/092002.
- UNITED NATIONS, “Concerning the Adoption of Uniform Technical Prescriptions for Wheeled Vehicles, Equipment and Parts which can be Fitted and/or be Used on Wheeled Vehicles and the Conditions for Reciprocal Recognition of Approvals Granted on the Basis of these Prescriptions,” Addendum 82: Regulation No. 83, Revision 5, (2015)
- ISO/TC 24/SC 4, “Aerosol particle number concentration – Calibration of condensation particle counters,” ISO 27891:2015, 2015.
- AEA Energy and Environment, “Particle Number Counter Calibration Procedures,” ED47382004/PNC Issue Number 5, December 2007.
- EUROPEAN COMMISSION, “COMMISSION REGULATION (EU) 2017/1154 of 7 June 2017 amending Regulation (EU) 2017/1151 supplementing Regulation (EC) No 715/2007 of the European Parliament and of the Council on type-approval of motor vehicles with respect to emissions from light passenger and commercial vehicles (Euro 5 and Euro 6) and on access to vehicle repair and maintenance information, amending Directive 2007/46/EC of the European Parliament and of the Council, Commission Regulation (EC) No 692/2008 and Commission Regulation (EU) No 1230/2012 and repealing Regulation (EC) No 692/2008 and Directive 2007/46/EC of the European Parliament and of the Council as regards real-driving emissions from light passenger and commercial vehicles (Euro 6),” (2017)
- Otsuki, Y., Kondo, K., Takeda, K., and Nakamura, H., “Applicability of Diffusion Charger Sensor to Portable Emission Measurement System,” SAE Technical Paper 2015-01-1994, 2015, doi:10.4271/2015-01-1994.
- Wilson, W.E., Stanek, J., Han, H.S., Johnson, T., et al.,, “Use of the Electrical Aerosol Detector as an Indicator of the Surface Area of Fine Particles Deposited in the Lung,” J. Air Waste Manag. Assoc. 57(2):211-220, 2007, doi:10.1080/10473289.2007.10465321.
- Kim, S., Kondo, K., Otsuki, Y., and Haruta, K., “A New On-Board PN Analyzer for Monitoring the Real-Driving Condition,” SAE Technical Paper 2017-01-1001, 2017, doi:10.4271/2017-01-1001.
- David B. Kittelson et al.., “Nanoparticle emissions from an ethanol fueled HCCI engine,” in 2010 Cambridge Particle Meeting, http://www.cambridgeparticlemeeting.org/sites/default/files/Presentations/2010/DKittelson(UMinnes ota)_2010_Ethanol_HCCI.pdf
- Otsuki, Y., Takeda, K., Haruta, K., and Mori, N., “A Solid Particle Number Measurement System Including Nanoparticles Smaller than 23 Nanometers,” SAE Technical Paper 2014-01-1604, 2014, doi:10.4271/2014-01-1604.
- Yoshinori Otsuki, Shigemi Tochino, Kenji Kondo and Kazuhiko Haruta, “Portable Emissions Measurement System for Solid Particle Number Including Nanoparticles Smaller than 23 nm,” SAE Technical Paper 2017-01-2402, 2017, doi:10.4271/2017-01-2402.