This content is not included in your SAE MOBILUS subscription, or you are not logged in.
Development of a Novel Ultrasonic Viscometer for Real Time and In-Situ Applications in Engines
ISSN: 0148-7191, e-ISSN: 2688-3627
Published April 14, 2015 by SAE International in United States
Annotation ability available
A novel ultrasonic viscometer for in-situ applications in engine components is presented. The viscosity measurement is performed by shearing the solid-oil contact interface by means of shear ultrasonic waves. Previous approaches to ultrasonically measure the viscosity suffer from poor accuracy owing to the acoustic miss-match between metal component and lubricant . The method described overcomes this limitation by placing an intermediate matching layer between the metal and lubricant. Results are in excellent agreement with the ones obtained with the conventional viscometers when testing Newtonian fluids. This study also highlights that when complex mixtures are tested the viscosity measurement is frequency dependent. At high ultrasonic frequencies, e.g. 10 MHz, it is possible to isolate the viscosity of the base, while to obtain the viscosity of the mixture it is necessary to choose a lower operative frequency, e.g. 100 kHz, to match the fluid particle relaxation time.
CitationSchirru, M., Sutton, M., Dwyer-Joyce, R., Smith, O. et al., "Development of a Novel Ultrasonic Viscometer for Real Time and In-Situ Applications in Engines," SAE Technical Paper 2015-01-0679, 2015, https://doi.org/10.4271/2015-01-0679.
- Kasolang, S., Ahmad, M.A., Dwyer-Joyce, R.S., “Measurement of circumferential viscosity profile in stationary journal bearing by shear ultrasonic reflection.”, J. Tribol. 133(3), 031501, June 17, 2011. doi:10.1016/j.triboint.2011.04.014.
- Mason, W.P., “Measurement of shear elasticity and viscosity of liquids at ultrasonic frequencies”, Physical Review, Volume 7S, Number 6, March I5 1949, pp 936-946. doi:10.1103/PhysRev.75.936
- Roth, W., “A New Method for Continuous Viscosity Measurement. General Theory of the Ultra-Viscoson”, Journal of Applied Physics, vol.24, No.7, July 1953, pp 940-950. doi:10.1063/1.1721406
- Sheen, S.H., Lawrence, W.P., Chien, H.T., Raptis, A.C., “Method for measuring liquid viscosity and ultrasonic viscometer”, Patent No: 5,365,778, Date of Patent: Nov. 22 1994.
- Cohen-Tenoudji, F., Ahlberg, L.A., Tittmann, B.R., Pardee, W.J., “High temperature ultrasonic viscometer”, Patent No.: 4,779,452, Date of Patent: Oct. 25 1988.
- Sheen, S.H., Chien, H.T., Raptis, A.C., “Ultrasonic methods for measuring liquid viscosity and volume percentage of solids”, ANL-97/4, Feb. 1997. doi:10.2172/477669
- Desilets, C.S., Fraser, J.D., Kino, G.S., “The design of efficient broad-band piezoelectric transducers”, IEEE Transactions On Sonics and Ultrasonics, Vol. SU-25, No. 3, May 1978. doi:10.1109/T-SU.1978.31001
- Harrison, G., Barlow, A. J., “Dynamic viscosity measurement”, Methods of Experimental Physics, Vol. 19, Ultrasonics. ISBN: 0-12-4755961-0
- Franco, E.E., Adamowski, J.C., Higuti, R.T., Buiochi, F., “Viscosity measurement of Newtonian liquids using the complex reflection coefficient”, IEEE Transactions on Ultrasonics, Ferroelectrics, And Frequency Control, Vol. 55, No. 10, Oct. 2008. doi:10.1109/TUFFC.923
- Greenwood, M.S., Bamberger, J.A., “Measurement of viscosity and shear wave velocity of a liquid or slurry for on-line process control”, Ultrasonics 39 (2002) 623-630. doi:10.1016/S0041-624X(02)00372-4
- Franco, E.E., Adamowski, J.C., Buiochi, F., “Evaluation of a mode conversion dual-cell to measure the complex reflection coefficient at a solid-liquid interface”, ABCM Symposium Series in Mechatronics, Vol. 4, pp.892-897, 2010.
- Buckin, V., Kudryashov, E., “Ultrasonic shear wave rheology of weak particle gels”, Advances in Colloid and Interface Science, 89-90 (2001), pp. 401-422. doi:10.1016/S0001-8686(00)00060-9