This content is not included in your SAE MOBILUS subscription, or you are not logged in.
Partial Flow Dilution System with Double Dilution for PM Sampling under Transient Test-Cycles
ISSN: 0148-7191, e-ISSN: 2688-3627
Published April 03, 2018 by SAE International in United States
This content contains downloadable datasetsAnnotation ability available
Automobile Particulate Matter (PM) Emission regulation requires gravimetric determination of PM collected on filter media under simulated driving conditions in the laboratory traditionally in a full flow Constant Volume Sampling (CVS) dilution tunnel. There have been discussions about whether current sampling and measurement practices are sufficiently accurate in quantifying PM at the upcoming 1mg/mi PM emissions standards of CARB LEV III. Sampling technique alternative to a CVS such as a Partial Flow Dilution (PFD) system has already been developed and is acceptable for certification testing. Lower dilution ratios and higher filter face velocity (FFV) are options to load traceable amount of PM on filter in case of light duty vehicle (LDV) testing. On the other hand higher dilution ratios and lower FFV are required for heavy duty engine (HDE) testing to keep the PM loaded on filter <400μg. A PFDS with a single dilution tunnel cannot support both LDV and HDE testing where a wide range of FFV and dilution ratios are required.
A new PFD system with double dilution has been developed in this study to meet PM measurement requirements of LDV and HDE transient testing. Two identical dilution tunnel configurations are employed for 1st stage and 2nd stage dilution. It uses fast response venturi flow meters that are free from errors caused by differences in the physical gas densities, or thermal capacity variations. The system can give minimum dilution ratio of 5:1 to maximum of 2500:1 and can operate under different control modes for dilution; proportional sampling mode and fixed dilution ratio mode. The PFD system developed in this study meets the sampling requirements of transient test cycles both for LDV and HDE with good flow accuracy, dilution ratio accuracy, and fast transient response for proportional sampling.
CitationRahman, M., Rooney, R., Nevius, T., and Otsuki, Y., "Partial Flow Dilution System with Double Dilution for PM Sampling under Transient Test-Cycles," SAE Technical Paper 2018-01-0643, 2018, https://doi.org/10.4271/2018-01-0643.
Data Sets - Support Documents
|Unnamed Dataset 1|
|Unnamed Dataset 2|
- Khalek , I.A. 2007
- Johnson , K.C. 2017
- Yamagishi , Y. and Ohtsuki , S.
- Odaka , M. A New Type Partial Flow Dilution Tunnel with Geometric Partitioning for Diesel Particulate Measurement SAE Technical Paper 2001-01-3579 2001 10.4271/2001-01-3579
- Schweizer , T. A New Approach to Particulate Measurement on Transient Test Cycles: Partial Flow Dilution as Alternative to CVS Full Flow System SAE Technical Paper 2000-01-1134 2000 10.4271/2000-01-1134
- Khalek , I.A. Performance of Partial Flow Sampling Systems Relative to Full Flow CVS for Determination of Particulate Emissions under Steady-State and Transient Diesel Engine Operation SAE Technical Paper 2002-01-1718 2002 10.4271/2002-01-1718
- Li , Y.
- Foote , E.B. Evaluation of Partial Flow Dilution Methodology for Light Duty Particulate Mass Measurement SAE Technical Paper 2013-01-1567 2013 10.4271/2013-01-1567
- Krenn , M. Full Flow and Partial Flow Dilution System for Emission Measurement-a Comparison regarding to SULEV Levels SAE Technical Paper 2001-01-3946 2001 10.4271/2001-01-3946
- Nevius , T. Techniques for Improved Correlation between Constant Volume and Partial Flow Sampling Systems SAE Technical Paper 2009-01-1351 2009 10.4271/2009-01-1351
- MacDonald , J.S. , Plee , S.L. , D’Arcy , J.B. , and Schreck , R.M. Experimental Measurements of the Independent Effects of Dilution Ratio and Filter Temperature on Diesel Exhaust Particulate Samples SAE Technical Paper 800185 1980 10.4271/800185
- Electronic Code of Federal Regulations Title 40